ATMON FL technical inspection
Professional, take care of your equipment!
A reminder! Calibration of the ATMON FL measuring modules should be performed every 6 months, but at least once a year!
All equipment requires maintenance and calibration - your ATMON FL does too!
Have you noticed something disturbing, or would you like to make sure everything is working properly and have ATMON FL serviced?
Request a technical inspection and send your ATMON FL to us. You don't have to worry about anything else. You will receive feedback on the result of the technical inspection. If ATMON FL does not require service, we will send it back to the address you have indicated.
If the inspection reveals a need for service work on your equipment, you will be informed. Before the NANOSENS service team carries out any additional work, a cost estimate will be sent to you for approval.
Maintenance of the PM module is carried out free of charge as part of the technical inspection service.
All technical inspections, maintenance and calibrations are carried out by the NANOSENS service, i.e. the manufacturer at its premises. This is carried out by a team of experienced service engineers, specialised in the construction and maintenance of gaseous substance measuring devices.
For the technical inspection it is advisable to order the calibration of the individual measuring modules that your ATMON FL is equipped with. Order here.
If you have any further questions, please contact us by e-mail - aeroMind team.
aeroMind - service ATMON FL
112B Piątkowska St.
Price includes packaging and return shipping.
If your Atmon FL is under warranty, a technical inspection is necessary to maintain it!
Detailed activities that make up the maintenance service:
(a) general overview of the technical condition of the device
b) evaluation of external mechanical damage, if any - before starting up the unit
c) checking the readings of the appliance after switching it on, before carrying out further service work
d) assessment of the condition of the appliance's internal modules (water, corrosion, wear and tear of
e) maintenance and cleaning of the above mentioned modules and elements
f) checking the operating parameters of the customer's power supply
g) checking the parameters of the power supply system, charging and battery consumption (full cycle
g) checking the parameters of the power supply, charging and battery consumption (full battery working cycle - 2x fully charged and fully discharged)
(h) status evaluation and technical parameter check of individual gas sensors
(i) software update of the control processor, if necessary
(a) checking the response of the sensor to the reference gases
(b) adjustment of the characteristics of the individual sensors
(c) final calibration of sensor sensitivity with calibration gas
3. replacement of the sensor: is carried out when the sensor is worn out or its parameters significantly
sensor is worn out or its parameters differ considerably from the required ones (e.g. due to working in the concentration range not intended
(a) Evaluation based on:
- initial analysis of sensor parameters as a result of inspection during the diagnostic process
- operating parameters of the sensor deviating from the standard in the sensor test carried out in the
the calibration process
b) Installation of a new sensor and carrying out stabilisation of the sensor (time for
stabilisation of the sensor parameters under operating conditions)
(c) checking the sensor parameters within the above mentioned period 1-3 times, depending on the sensor
(d) calibrating with reference gases
4. service repair of faults detected in the diagnostic process other than:
(a) sensor wear or damage
(b) soiling requiring standard cleaning without the use of additional cleaning and maintenance agents
cleaning and maintenance agents
The repair shall be commenced upon the acceptance by the customer of the submitted repair costs
Each service repair invoice is accompanied by a service protocol specifying the scope of the performed actions, used elements, subassemblies.